Pushing any Operations key will immediately restart tester
and will supersede prior command.
D - DATA
Displays Density count.
M - DATA
Displays Moisture count.
D - STD
Displays Density standard count.
M - STD
Displays Moisture standard count.
Upon initial operation after a period of nonuse, press the
During the count cycle, a pulsing "beep" will be heard
CLEAR key to clear out any unwanted commands.
If the batteries are low, but still useable, an "L" will
Press any of the four OPERATIONS keys to start the
appear in the display window. If the batteries are below
tester for normal density and moisture counting. All
acceptable charge level, the "L" will display, but the
displays will be normalized to a 1/4 minute count base for
tester will "Cutout" and will not operate. Recharge the
simplicity of computation.
tester overnight when the batteries indicate "L" or low.
At the completion of any count cycle, the tester will
"beep-beep" rapidly for ten beeps.